Download the newest software version to a USB storage device. After downloading is complete, remove the USB from your computer. Insert the USB into one of the USB ports of your QuikRead go Instrument and start Settings - Measurement flow - Maintenance - Software update.
About the USB storage device:
Size of the USB storage device should be 1 GB or larger, it should be in FAT32 format and partitioned as a primary partition, marked as active. Before downloading the new software version into the USB storage device, please make sure that your QuikRead go Instrument detects the USB sorage device. You can do it by inserting the USB to one of the QuikRead go Instrument's USB-ports. A USB icon will appear on the display when detection is ready.
Extracting the compressed content:
1. The downloaded files are compressed in an archive folder and need to be extracted to the USB storage device. Navigate to your Downloads folder where you will find the downloaded software folder. Different browsers may have slightly different commands and names for archive folders.
Note: Don't change any of the file or folder names when downloading the files to the USB. The name of the folder should be "qrzwupdate".
2. Right click the downloaded folder and select "Extract all”. Select your USB and extract the content there. “Extract all” -command can also be found in: Open with -> 7-zip.
you can find a more detailed instruction for extracting the files.
How to upload the newest software to your QuikRead go instrument:
How many times can I download a software version for one QuikRead go instrument?
Please fill in the region, serial number and user information as required above.
You can download a newest software version twice for a same instrument serial number.
The count will always be reset when a new software version is published and uploaded to the site, so you will be able to use the same serial number again.